Comparison of the Heat Generation of Light Curing Units
Bora Bağiş, Yildirim Bagis, Ertan Ertas, Seda Ustaomer
Citation Information :
Bağiş B, Bagis Y, Ertas E, Ustaomer S. Comparison of the Heat Generation of Light Curing Units. J Contemp Dent Pract 2008; 9 (2):65-72.
CC BY-NC 3.0
Copyright © 2008; The Author(s).
The aim of this study was to evaluate the heat generation of three different types of light curing units.
Methods and Materials
Temperature increases were recorded from a distance of 1 mm from a thermocouple to the tip of three different types of light curing units including one quartz-tungsten halogen (QTH), one plasma arc (PAC), and one light emitting diode (LED) unit. An experimental model was designed to fix the 1 mm distance between the tip of the light curing units and the thermocouple wire. Temperature changes were recorded in 10 second intervals up to 40 seconds. (10, 20, 30, and 40 seconds). Temperature measurements were repeated three times for every light curing unit after a one hour standby period. Statistical analysis of the results was performed using the analysis of variance (ANOVA) and the Bonferroni Test.
The highest temperature rises (54.4±1.65°C) occurred during activation of a PAC light curing unit for every test period (p<.05). The least temperature increase (11.8±1.3°C) occurred with a LED curing unit for each tested period except for the measurement of the temperature rise using the QTH curing unit at the tenth second interval (p<.05).
These results indicate the choice of light activation unit and curing time is important when polymerizing light activated resin based restorations to avoid any thermal damage to the pulp.
Bagis B, Bagis Y, Ertas E, Ustaomer S. Comparison of the Heat Generation of Light Curing Units. J Contemp Dent Pract 2008 February;(9)2:065-072.
In vitro pulp chamber temperature rise during composite resin polymerization with various light-curing sources. Dent Mater. 1999; 15:275-281.
Temperature rise during polymerization of light-activated resin composites. J Oral Rehab. 1998;25:908–913.
Polymerization and light-induced heat of dental composites cured with LED and halogen technology. Biomater. 2003;24:1809–1820.
Influence of heat from light curing units and dental composite polymerization on cells in vitro. J Dent. 2006 Apr;34(4):298-306.
Thermogenesis in operative techniques, comparison of four methods. J Prosthet Dent. 1962;12:977–984.
Microscopic observations on living dental pulp. Acta Odontol Scand. 1958;16:303–327.
The effect of different light-curing units on fatigue behavior and degree of conversion of a resin composite. Dent Mater. 2005; 21:608-615.
Thermal risks from LED- and high-intensity QTH-curing units during polymerization of dental resins. Biomed Res. 2005; 72:260-267.
Thermal changes and cure depths associated with a high intensity light activation unit. J Dent. 2005; 32;643-651.
Thermographic measurement of temperature change during resin composite polymerization in vivo. J Dent. 1995; 23:267-271.
Biological effects of blue light from dental curing units. Dent Mater. 2004; 20:150-157.
Temporary increase during resin cement polymerization under ceramic restoration: Effect type of curing unit. Int J Prosthodont. 2004; 17;200-204.
Advances in light curing. J Dent. 2000;13:77–81.
Light emitting diode (LED) polymerization of dental composites: flexural properties and polymerisation potential. Biomater. 2000;21:1379–85.
Orthodontic Bracket Shear Bond Strengths Produced by Two High-power Light-emitting Diode Modes and Halogen Light. Angle Orthodont. 2005;75:854–857.
Chemistry of synthetic resins. In: Phillips’ science of dental materials. 10th ed. Philadelphia: WB Saunders, 1996; 216.
Council on Dental Materials, Instruments, and Equipment. JADA. 1985;110:100–23.
Comparison of polymerization by light-emitting diode and halogenbased light-curing units. Am J Dent Assoc. 2002; 133:335-341.
Pulpal temperature rise during light-activated bleaching. J Biomed Mater Res B Appl Biomater. 2005; 72;254-259.
Temperature Rise During Orthodontic Bonding With Various Light-curing Units—An Invitro study. Angle Orthodont. 2004;76:330–334.
Dental composite depth of cure with halogen and blue light emitting diode technology. Br Dent J. 1999;186:388–91.
Depth of cure and compressive strength of dental composites cured with blue light emitting diodes (LEDs). Dent Mater. 2000;16(1):41–7.
Lifetime intensity profiles of light-curing units. J Dent Res. 1996; 75:380.
Curing of dental composites by use of InGaN light-emitting diodes. Opt Lett. 1999; 24:67-68.
Performance of two blue light-emitting-diode dental light curing units with distance and irradiation-time. Dent Mater. 2004; 20:72-79.
Candela-class high brightness InGaN/AlGaN double heterostructure blue-light-emitting diodes. Appl Phys Lett 1994;64:1687–9.
Handbook of optics. Vol. 2: Devices, measurements, and properties. New York: McGraw-Hill, 1995;121–123.
An in vivo study to compare a light and a conventional quartz halogen curing light in ortodontiv bonding. Eur J Orthod. 2004; 26:573-577.
Huck, Kahl-Nieke B. Curing lights for orthodontic bonding. Am J Orthod Dentofacial Orthop. 2002; 122:643-8.
Effects of conventional and high-intensity lightcuring on enamel shear bond strength of composite resin and resin-modified glass-ionomer. Am J Orthod Dentofacial Orthop. 2000; 119:30-5.
Characterization of resin composites polymerized with curing units. Dent Mater. 2000;16:330–336.
Cure performance of light-activated composites by differential thermal analysis (DTA). Dent Mater. 1985;1:231–234.
Pulp chamber temperature changes with visible-lightcured composites in vitro. Dent Mater. 1990;6:99–102.
Computer-controlled differential calorimetry of dental composites,. IEEE Trans Biomed Eng. 1991;131:319–325.
Thermographic investigation of contemporary resin-containing dental materials. J Dent. 2005; 33:593-602.
In vitro temperature rises produced by five polymerising light sources. Restorative Dent. 1988;4:33–35.
The use of differential scanning calorimetry for the evaluation of dental materials. I. Cements, cavity lining materials and anterior restorative materials. J Oral Rehabil. 1980;7:103–110.
Effect of Light-tip Distance on the Shear Bond Strengths of Composite Resin. Angle Orthodont 2005;75:386–391.
Comparison of quartz-tungsten-halogen, light-emitting diode, and curing lights. J Adhes Dent. 2003; 5(3):193-207.
Effect of different light sources and guides on shear bond strength of brackets bonded with 2 adhesive systems. Am J Orthod Dentofacial Orthop. 2005; 128:99-102.
Thermal emission by different light-curing units. Oper Dent. 2003;28:260–266.
Jandt Polymerization and light-induced heat of dental composites cured with LED and halogen technology. Biomater. 2003; 24:1809-1820.
Temperature rise in the pulp chamber induced by a conventional halogen light-curing source and a lamp. Am J Dent. 2004;17:203-8.